not necessary, since there is no fractional part of a space to determine.

The reading is 2.350. The 0 mark on the Vernier scale, as shown in figure

188, view B, on the previous page, coincides with a fractional part of a

space on the rule. The reading is 2.35 plus a fraction of the space on the

rule. In order to determine what fractional part of a whole rule division,

or how many thousands are to be added to the 2.35 reading, it is necessary

to find the line on the Vernier scale that exactly coincides with the line

on the rule.

In this case, the line coincides at the 18th mark.

This

indicates 18/25 of a whole space. Since each space on the rule equals 0.025

inch, this part of a space is equal to 0.018 inch, and the total reading is

2.34 plus 0.018, or 2.368 inches.

Vernier scales are not necessarily 25 divisions long;

they may have any number of units. For example, the

ten thousandths micrometer has a Vernier scale of only

ten divisions.

(b) *Applications. * The Vernier caliper has a wide range of measurement

applications, and the shape of the measuring jaws and their position with

respect to the scale makes this tool more adaptable than a micrometer.

However, the Vernier caliper does not have the accuracy of a micrometer. In

any 1 inch of its length, a Vernier caliper should be accurate within 0.001

inch.

In any 12 inches, it should be accurate within 0.002 inch and

increase about 0.001 for every additional 12 inches.

The accuracy of

measurements made with a Vernier caliper is dependent on the user's ability

to feel the measurement. Because the jaws are long, and because there is

the possibility of some play in the adjustable jaw, especially if an

excessive measuring pressure is used, it is necessary to develop an ability

to handle the Vernier caliper. This touch may be acquired by measuring such

known standards as gage blocks and plug gages. Applications of the Vernier

caliper are shown in figure 189 on the following page.

In view A, a

machinist is checking the outside diameter of a part. One hand holds the

stationary jaw to locate it, while the other hand operates the adjusting nut

Integrated Publishing, Inc.
6230 Stone Rd, Unit Q
Port Richey, FL 34668
Phone For Parts Inquiries: (727) 493-0744 Google + |