OD1402 - Lesson 1/Learning Event 1
o Turn the LEVEL knob to midrange.
o Turn the SLOPE knob fully clockwise to +.
o Set the MODE SWITCH to AUTO.
o Set the COUPLING switch to AC.
o Set the SOURCE switch to INT.
o Set the MAG switch to xl by depressing the button.
The TIME/DIV OR DLY TIME switch is four controls in one.
Care must be taken to see that each is positioned
o Set the innermost TIME/DIV switch to the 1ms position.
o Set the DLY. TIME switch to the 1ms position and then push
o Turn the VARIABLE (CAL) knob fully clockwise until it clicks
o Turn the DLYD TRG knob fully clockwise to RUNS AFTER DLY
o Set the SLOPE to +, COUPLING to AC, and SOURCE to INT by
depressing the pushbuttons.
o Turn the DELAY TIME MULT knob to the 0 position.
So far, the basic controls of the oscilloscope and their functions
have been discussed, as well as the initial setup of those
controls that must be accomplished prior to turning on the scope.
You are now ready to turn on the oscilloscope.
In the next learning event, how to turn on and adjust the
oscilloscope for a single trace will be discussed. Then, you will
learn how to measure the voltage amplitude and frequency of a
signal by using the screen grid.