probably open" not "DC voltmeter bad." By being specific they can assist repair

personnel and insure rapid return of the equipment to a serviceable condition.

Figure 9.

DC voltmeter.

f. Figure 10 shows the simplified diagram of a typical DC ammeter. The ammeter

circuit must shunt most of the current being measured away from the meter movement.

On the 10 milliampere (10mA) range, the meter movement resistance of 2K ohm is in

series with the 3K ohm resistance of R11. R17, R18, R19 and R22 are in series with

each other and are in parallel with R11 and the meter movement. Resistive values

have been selected so that 50A will flow through the meter and R11 and 9.950mA

will flow through R22, R17, R19 and R18 when 10mA is being measured. The truth of

this statement can be proved using Ohm's law. Let "x" be equal to the amount of

current flowing through the meter when there is a 10mA current input to the

circuit. The voltage across both parallel sections of the circuit must be equal so:

(1) x(2K 3K) = (.01 -X)(.025 .475 2 22.5)

(2) x(5K) = (.01 -X)(25)

(3) 5000X = .25 - 25X

(4) 5025X = .25

(5)

(6) x = .0000497 ampere or 49.7A

Ohm's law proves the validity of the circuit; 49.7A is close enough to 50A to

produce the required accuracy.

h. When the circuit selector switch is moved from the 10mA to the 100mA

position, R18 is placed in series with R11 and the meter movement. This circuit

change will allow a full scale deflection on the meter to be representative of a

100mA input current. Using Ohm's law, calculate the meter current at full scale

under these circuit conditions. Again, your answer should be 49.7A.

OS 010, 7-P13

Integrated Publishing, Inc.
6230 Stone Rd, Unit Q
Port Richey, FL 34668
Phone For Parts Inquiries: (727) 493-0744 Google + |